Minerva patient lift recall
This article was originally published in The Gray Sheet
Medibo initiates Class I recall of 64 units of the battery-powered patient lifts Aug. 4 due to mechanical problems. Some hanger bars detached because of missing spring washers, causing patients to fall, the agency says. Further Minerva problems cited include loose foot-pedal assembly bolts and faulty actuator brackets on the mast assembly, resulting in lift instability. The recall continues a rising trend in Class I recalls (1"The Gray Sheet" July 19, 2004, p. 16). The Minerva patient lift is technically identical to Arjo's Minstrel, recalled earlier this year...
You may also be interested in...
Carl Zeiss Meditec's Class I voluntary recall is part of a recent uptick in FDA announcements related to the most serious type of device recalls
As the first emergency use authorization is granted, Pfizer and Moderna stand accused of neglecting poorer nations as wealthy countries snap up early access to COVID-19 vaccines.
The FDA has released final guidance on “Formal Meetings Between FDA and ANDA Applicants of Complex Products Under GDUFA” for the industry. The guidance brings clarity and aims at facilitating product development meetings between the FDA and the ANDA applicants for complex products.