Minerva patient lift recall
This article was originally published in The Gray Sheet
Medibo initiates Class I recall of 64 units of the battery-powered patient lifts Aug. 4 due to mechanical problems. Some hanger bars detached because of missing spring washers, causing patients to fall, the agency says. Further Minerva problems cited include loose foot-pedal assembly bolts and faulty actuator brackets on the mast assembly, resulting in lift instability. The recall continues a rising trend in Class I recalls (1"The Gray Sheet" July 19, 2004, p. 16). The Minerva patient lift is technically identical to Arjo's Minstrel, recalled earlier this year...
You may also be interested in...
Carl Zeiss Meditec's Class I voluntary recall is part of a recent uptick in FDA announcements related to the most serious type of device recalls
While endocrine-disrupting evidence was inconclusive, the Scientific Committee on Consumer Safety recommends more conservative limits on use of homosalate, octocrylene and benzophenone-3 in cosmetic products compared with current requirements under the European Cosmetics Product Regulation.
The risk of inadvertently growing SARS-CoV-2 virus in cell and gene therapies and possibly infecting patients and workers should be assessed and mitigated, the agency advises.